Characteristics: It is the basic surgical equipment, which can effectively cut all kinds of bandages. It uses high-speed saw blade vibration system. It will not scrach the skin for the blade rolls from left to right with the amplitude of 5.6.
How to use: Grasp the electric plaster saw and the cutting blade should be vertical on the cast's surface and make stress slowly. when the material was cut through, put up the saw blade slightly and move forward along the incision. Repeat the above process until the material completely cut open
Product Registration No.: (Z) 2013No. 1100005Product Standard No.: YZB / Su (Yan) 0006 - 2013
Electric Plaster Saw Instructions
Components and Uses
Electric Plaster Saw consists of many components, including electric motor, propeller shaft, spindle sleeve, eccentric shaft, axletree, transmission pliers, circular saw blade, line shape blade and casing, preoperculum, pinching screw, power cord, wire plug and so on.
This product is mainly used for patients who recovery from orthopaedics operations. It is the basic surgical equipment which can effectively cut all kinds of bandages.
Structural Characteristics and Principle of Work
Via eccentric shaft driving transmission pliers, the electric motor is driven by electrical source, then it converted the blade into oscillates traverse motion and saws the bandage. No injuries happened during bandage removing.
Electric motor adopts double insulated protection device to ensure safety for the operators and patients. The characters of Electric Plaster Saw cover the following:
1. Powerful motion
2. Ingenious design
3. High operation stability and amplitude
4. High safe reliability
5. Low noise
Running Conditions and Essential Parameter
1. Normal running conditions:
Ambient temperature: 5~40℃
Relative humidity: ≤80%
Supply Voltage: 220V±22V
Power frequency: 50HZ±1HZ
2. Essential parameter:
a). Oscillative angle: 6º±1º
b). Oscillative frequency (running-idle): 15000times/min±2250times/min
c). Noise of idle load: ≤90dB(A)
d). Temperature range of idle load: the temperature of machine casing is no more than 25℃ after running 3 minutes later.
1. Before using, to check whether the saw bit is smooth, whether missing sawtooth, whether installation is tight. It must be tightened with 5mm inner hexagon spanner in a clockwise direction when installing.
2. Before the operation, to connect electric motor power and then switch on equipment after disinfecting the electric plaster saw.
3. Grasp the electric plaster saw and the saw blade should be vertical on the bandage’s surface where need to be removed and make stress slowly, then cut the bandage vertically. Please don’t cut the bandage at one time, although the saw has function of soft-tissue protection.
4. To cut the vertical surface of the bandage, and then move forward slowly to the place where need to be removed. And repeat this procedure until completion.
5. After long-term usage, the sawtooth may become blunt. It just needs to release the screw cap with 5mm inner hexagon spanner in a counterclockwise direction. And then tighten it in a clockwise direction after wheeling the saw bit,
1. Wire Plugs should be installed in three sockets with the same specification. To ensure safe use of electricity, the secure socket should be grounded.
2. To inspect whether saw bit is smooth, whether missing sawtooth, whether installation is tight before using.
3. To exert proper pressure instead of excessive pressure while the saw bit oscillating
4. Do cut off the electrical source and withdraw the saw bit slowly while stuck in the bandage. Do not switch on and off the switch alternatively which will damage electric motor under such condition of changeful charge load.
5. The temperature of equipment shell will increase with the using time goes by. It should be paused a moment if discover the temperature is overheating. Restart it while cooling. Otherwise, the electrical motor would be damaged by overheating.
6. If there is an abnormal sound, too larger vibration or load reduction, etc., it should be promptly sent to manufacturers or electricians for inspection and repair.
Care and Maintenance
1. To inspect insulating property of electrical motor regularly to ensure the safe use of electricity.
2. Inspect all parts at regular intervals. To inspect the mechanical structure if the motor is operating normally but parts are at fault, then change accessories or repair according detailed situation. In order to avoid driving medium worn out and affecting using life due to lacking of oil, adding MOS2 grease # 1 after repairing.
Commitment of After-Sales Service
We hereby declare seriously the product which caused by quality problem can be serviced for free within half a year. We also declare that only material cost collected during On the occasion of one-year-warranty period.
Beijing Jinwei Kangda Medical Instrument Ltd. was founded in 2001, which specialize in these products’ research and development, production, sales at home and abroad. We are the supplier of the central materials reserve, China MEHOCO CO., LTD’s export and military material procurement. The company has 20 production lines of body protector, orthosis brace. Jinwei Kangda as the high-tech enterprises has won a number of national patents and quality excellent award of Beijing. All the products have ISO, CE and FDA certificates. Our products are used in thousands of hospitals in China and exported to Europe, Middle East and Southeast Asia and more than 30 countries, we are the largest base of orthopedic casting tape and splint research, manufacture, exportation in Asia.
Sincerely welcome companies at home and abroad contact and cooperate with us to realize complementary advantages, win-win cooperation and harmonious development together. Company address: Room 1406-1409, 1st Building, No.14 Courtyard, South Road of Western 3rd Ring-Road, Fengtai District, Beijing 100073, China.
1. Free sample.
2. Small order acceptable.
3. FOB, CNF, CIF, EXW terms available.
4. 3rd part product inspection is acceptable.
5. Manufacturer with competitive and stable price.
6. OEM & ODM service with more options on color, size and packing etc.
7. Shipping and express will be tracked and reported at any time after sending.
8. Good after-sale service, we take care complaints as we want to be your long-term supplier.
All the products are approved by CE, ISO9001, ISO13485 and FDA.
Supplier of Chinese Army and government.
3 years of warranty period with reliable quality, accept changing of goods with problems any time, confidence in quality because of 15 years of professional manufacturing and exportation, over 200 experienced workers and management.
The factory covers an area of 28,600 square meters and 11,600 square meters building area with 24 production lines of casting tape and splint, produce over 10 millions of products each year.
Export to over 30 countries in Europe, Asia, and North America etc for 15 years.